Posted On :

The static characteristics of measuring instruments are concerned only with the steady-state reading that the instrument settles down to, such as accuracy of the reading.

**STATIC AND DYNAMIC
RESPONSE **

The static characteristics of measuring
instruments are concerned only with the steady-state reading that the
instrument settles down to, such as accuracy of the reading.

The dynamic characteristics of a
measuring instrument describe its behavior between the time a measured quantity
changes value and the time when the instrument output attains a steady value in
response. As with static characteristics, any values for dynamic
characteristics quoted in instrument data sheets only apply when the instrument
is used under specified environmental conditions. Outside these calibration
conditions, some variation in the dynamic parameters can be expected.

In
any linear, time-invariant measuring system, the following general relation can
be written between input and output for time (t) > 0:

where
q* _{i}* is the measured quantity, qo is the output reading, and a

**Zero-Order Instrument **

If
all the coefficients a1 . . . an other than a0
in Equation (2) are assumed zero, then where K is a constant known as
the instrument sensitivity as defined earlier. Any instrument that behaves
according to Equation (3) is said to be of a zero-order type. Following a step
change in the measured quantity at time t, the instrument output moves
immediately to a new value at the same time instant t, as shown in Figure. A
potentiometer, which measures motion is
a good example of such an instrument, where the output voltage changes instantaneously
as the slider is displaced along the potentiometer track.

**First-Order Instrument **

If all the coefficients a2 . . . an except for ao
and a1 are assumed zero in Equation (2) then

Any instrument that behaves according to Equation
(4) is known as a first-order instrument. If d/dt is replaced by the D operator
in Equation (4), we get

Defining K ¼ b_{0}/a_{0} as the
static sensitivity and t ¼ a_{1}/a_{0} as the time constant of
the system,

Equation (5) becomes

**Second-Order Instrument
**

If all coefficients a_{3} . . . other than a_{0},
a_{1}, and a_{2} in Equation (2) are assumed zero, then we get

This is the standard
equation for a second-order system, and any instrument whose response can be
described by it is known as a second-order instrument. If Equation (9) is
solved analytically, the shape of the step response obtained depends on the
value of the damping ratio parameter x. The output responses of a second-order
instrument for various values of x following a step changein the value of the
measured quantity at time t are shown in Figure.
Commercial second-order instruments, of which the accelerometer is a common
example, are generally designed to have a damping ratio (x) somewhere in the
range of 0.6–0.8.

Tags : Mechanical - Metrology and Measurements - Concepts of Measurement

Last 30 days 181 views
Related words : ### What is Static and Dynamic Response Define Static and Dynamic Response Definition of Static and Dynamic Response where how
meaning of Static and Dynamic Response
lecturing notes for Static and Dynamic Response lecture notes question and answer for Static and Dynamic Response answer
Static and Dynamic Response study material Static and Dynamic Response assignment Static and Dynamic Response reference description of Static and Dynamic Response
explanation of Static and Dynamic Response brief detail of Static and Dynamic Response easy explanation solution Static and Dynamic Response wiki
Static and Dynamic Response wikipedia how why is who were when is when did where did Static and Dynamic Response list of Static and Dynamic Response school assignment college assignment
Static and Dynamic Response college notes school notes kids with diagram or figure or image difference between Static and Dynamic Response www.readorrefer.in - Read Or Refer

Recent New Topics :
| Performance of Wind Machines | | Analysis of Aerodynamic Forces Acting on the Blade | | Structure and components of wind mill | | Savonius wind turbine: Schematic Structure, Advantage, Disadvantages | | Important Short Questions and Answers: Solar Radiation and Solar Energy Collectors | | Flat Plate Collector Efficiency Factor | | Flat Plate Collectors: Principle, Components, Types | | Solar Radiation Data | | Solar Radiaion Geometry | | Solar Radiation at the Earthâ€™s Surface | | Solar Constant | | Solar Spectrum | | Solar Radiation and Solar Energy Collectors | | Important Short Questions and Answers: Industrial Heating and Welding | | Laser Beam Welding | | Welding Transformer | | Welding Generator - A.C Supply | | Arc welding | | Resistance welding | | Types of electric welding | | Introduction to electric welding | | Electric arc furnaces: Advantages, Application | | Dielectric heating: Advantages, Application | | Induction heating: Types, Advantages, Application | | Resistance Heating: Types, Advantages, Application |