Posted On :

Sine bars are always used along with slip gauges as a device for the measurement of angles very precisely. They are used to 1) Measure angles very accurately. 2) Locate the work piece to a given angle with very high precision.

**SINE BAR **

Sine bars are always used along with
slip gauges as a device for the measurement of angles very precisely. They are
used to

1) Measure
angles very accurately.

2) Locate
the work piece to a given angle with very high precision.

Generally, sine bars
are made from high carbon, high chromium, and corrosion resistant steel. These
materials are highly hardened, ground and stabilized. In sine bars, two
cylinders of equal diameter are attached at lie ends with its axes are mutually
parallel to each other. They are also at equal distance from the upper surface
of the sine bar mostly the distance between the axes of two cylinders is 100mm,
200mm or 300mm. The working surfaces of the rollers are finished to 0.2µm
R value. The cylindrical holes are provided to reduce the weight of the sine
bar.

**Working principle of
sine bar**

The working of sine bar
is based on **trigonometry principle**. To measure the angle of a given
specimen, one roller of the sine bar is placed on the surface plate and another
one roller is placed over the surface of slip gauges. Now, ‘h be the he slip
gauges and ‘L’ becenters,thenthe angledistancecalculatedas between

**Use of Sine Bar **

**Locating any’
work to a
given angle**

1) Before checking the unknown angle of the
specimen, first the angle (0) of given specimen is found approximately by bevel
protractor.

2) Then
the sine bar
is set at
angle of 0
and clamped on the angle plate.

3) Now, the work is placed on the sine bar
and the dial indicator set at one end of the work is moved across the work
piece and deviation is noted.

4) Slip gauges are adjusted so that the dial
indicator reads zero throughout the work surface.

**Limitations
of sine bars**

1) Sine
bars are fairly reliable for angles than 15°.

2) It
is physically difficult to hold in position.

3) Slight
errors in sine bar cause larger angular errors.

4)
A difference of deformation occurs at
the point of roller contact with the surface plate and to the gauge blocks.

5) The
size of parts to be inspected by sine bar is limited.

**Sources
of error in sine bars**

The
different sources of errors are listed below:

1) Error
in distance between roller centers.

2) Error
in slip gauge combination.

3) Error
in checking of parallelism.

4) Error
in parallelism of roller axes with each other.

5) Error in flatness of the upper surface of sine bar.

Tags : Mechanical - Metrology and Measurements - Linear and Angular Measurements

Last 30 days 643 views
Related words : ### What is Sine bar Define Sine bar Definition of Sine bar where how
meaning of Sine bar
lecturing notes for Sine bar lecture notes question and answer for Sine bar answer
Sine bar study material Sine bar assignment Sine bar reference description of Sine bar
explanation of Sine bar brief detail of Sine bar easy explanation solution Sine bar wiki
Sine bar wikipedia how why is who were when is when did where did Sine bar list of Sine bar school assignment college assignment
Sine bar college notes school notes kids with diagram or figure or image difference between Sine bar www.readorrefer.in - Read Or Refer

Recent New Topics :
| Important Short Questions and Answers: Industrial Heating and Welding | | Welding Transformer | | Welding Generator - A.C Supply | | Arc welding | | Resistance welding | | Resistance Heating: Types, Advantages, Application | | Classification of Methods of Electric Heating | | Mode of Heat Transfer | | Advantage of Electric Heating | | Requirement of Heating Material | | Role electric heating for industrial applications | | Electric Heating Introduction | | Important Short Questions and Answers: Illumination | | Energy Efficient Lighting : LED and CFL | | Sports Lighting | | Interior Lighting | | Glare Evaluation | | Flood Lights | | Road Lighting | | Industrial Luminaries | | Types of Lighting | | Design of lighting system | | Lighting Calculations | | Fluorescent Lamp | | Sodium Vapour Lamp |