Posted On :

In the method of repetition, the horizontal angle is measured a number of times and the average value is taken.

**Method of Repetition in Theodolite**

In the method of repetition, the horizontal angle is measured a
number of times and the average value is taken. It is usual to limit the number
of repetitions to three with each face except in the case of very precise work.
With large number of rep-etitions, errors can also increase due to bisections,
reading the verniers, etc. Very large number of repetitions necessarily do not
lead to a more precise value of the angle. However, a number of errors are
eliminated by the repetition method. The procedure is as follows (Fig. 6.6).

1. Angle POQ is to be measured. Set up, centre, and level the
theodolite at O. Ensure that the instrument is in the normal position, i.e.,
face left.

2. Set the instrument to read 0� 00� 00?. For this release the upper
clamp and bring the zero of the vernier (at vernier A) very close to the zero
of the circle. Clamp the upper plate and using the upper tangent screw,
coincide the two zeros exactly.

3. Loosen the lower clamp and rotate the instrument so that the left
signal at P is approximately bisected. Tighten the lower clamp and using the
lower tangent screw, bisect the signal at P exactly. Read the verniers at A and
B. The reading should not change and they should read zero and 180�.

4. Loosen the upper clamp and rotate the instrument clockwise to bisect
the right signal at Q. Using the upper tangent screw, biscect the signal at approximately
Q exactly.

5. Read the verniers at A and B. The reading at A gives the value of the angle directly. The reading on the vernier at B will be 180� + the angle. Record both the readings.

Release the lower clamp and rotate the instrument clockwise to bisect the sig- mal at the left station P again. Using the lower tangent screw, bisect the signal

The method
of repetition helps to eliminate the following errors.

(a) Errors caused by the eccentricity of the centres and verniers, by
reading both the verniers and averaging.

(b) Graduation errors by reading from different parts of the circle.

(c) Imperfect adjustment of the line of collimation and horizontal
axis by face-left and face-right observations.

(d) Observational errors and other errors tend to be compensated by
the large number of readings.

However, the errors due to levelling
cannot be compensated. This has to be done by permanent adjustment. Also a
large number of repetitions tend to increase the wear of clamp and tangent
screws.

Therefore,
from the two sets,

Mean value of the angle = (1/2)(61� 42� 47? + 61� 42� 40? ) = 61�
42� 44?

Tags : Civil Surveying - Theodolite Surveying

Last 30 days 130 views
Related words : ### What is Method of Repetition in Theodolite Define Method of Repetition in Theodolite Definition of Method of Repetition in Theodolite where how
meaning of Method of Repetition in Theodolite
lecturing notes for Method of Repetition in Theodolite lecture notes question and answer for Method of Repetition in Theodolite answer
Method of Repetition in Theodolite study material Method of Repetition in Theodolite assignment Method of Repetition in Theodolite reference description of Method of Repetition in Theodolite
explanation of Method of Repetition in Theodolite brief detail of Method of Repetition in Theodolite easy explanation solution Method of Repetition in Theodolite wiki
Method of Repetition in Theodolite wikipedia how why is who were when is when did where did Method of Repetition in Theodolite list of Method of Repetition in Theodolite school assignment college assignment
Method of Repetition in Theodolite college notes school notes kids with diagram or figure or image difference between Method of Repetition in Theodolite www.readorrefer.in - Read Or Refer

Recent New Topics :
| Business Television (BTV) - Adaptations for Education | | System reliability and design lifetime | | Inter modulation and interference | | Downlink | | The Uplink | | Thermal Control and Propulsion | | Attitude Control & Orbit Control | | The Power Supply- Satellite | | Spacecraft Technology- Structure | | Important Questions and Answers: Satellite Orbits | | Launching Procedures | | Eclipse | | Geo stationary and Non Geo-stationary orbits | | Orbital Parameters | | Newton’s law | | Kepler’s laws | | Q and Phase Shift | | Measurement of SWR and Attunuation | | Measurement of Wavelength and Impeadence | | Important Short Questions and Answers: Microwave Passive Components | | Important Questions and Answers: RF Transistor Amplifier Design and Matching Networks | | Important Short Questions and Answers: Recent Trends in Medical Instrumentation | | Electrical Safety of Medical Equipment | | Diathermy Units | | Lasers in Medicine |